We use these bridges for measurement of inductances and capacitances in the most accurate manner. Principally, an AC Bridge is similar to the Wheatstone bridge. Like Wheatstone Bridge, it has also four arms. Instead of using pure resistances in the arms, we use impedances in the case of an AC Bridge.
Now suppose, Z1, Z2, Z3, and Z4 are the impedances of four arms of the bridge. Instead of using a DC galvanometer in the case of Wheatstone Bridge, we use here vibration galvanometer or detector. Instead of a battery, we use here an alternating source of voltage.
Theory of AC Bridge
Before going to the actual theory of AC Bridge, let us recall the theory of Wheatstone bridge.
As shown in the figure above, at the balanced condition of the bridge,
In this bridge, we use impedances instead of resistances. The balanced condition of an AC Bridge is the same as that of the Wheatstone Bridge. That means here we will write,
The impedance has two components in its polar form. One is its magnitude and the other is its phase angle. So, we can write the expression of the impedances, as |Z1|∠φ1, |Z2|∠φ2, |Z3|∠φ3 and |Z4|∠φ4. Now, we can write the expression of the balanced condition of the bridge circuit as,
From the above expression, we can conclude that there would be two conditions, for a balanced AC bridge with four impedance arms. Obviously, the balanced bridge must satisfy these two conditions simultaneously. And these conditions are
There are many types of AC bridges. We use them for different measurements, such as
For measuring self-inductances
For measuring capacitances
- AC Bridges and Theory of AC Bridge
- Maxwell Bridge Construction and Theory of Maxwell Bridge
- Maxwell Wien Bridge Circuit and Theory
- Hay Bridge Circuit Construction and Theory of Hay Bridge
- Murray Loop Test for Finding Fault Location in Cable
- Anderson’s Bridge Circuit and Theory of Anderson Bridge
- Wheatstone Bridge Circuit Operation and Theory
- Schering Bridge Construction and Theory
- De Sauty Bridge Construction Circuit and Theory
- Heaviside Campbell Equal Ratio Bridge
- Owen Bridge Circuit and Theory of Owen Bridge
- Varley Loop Test for Finding Fault Location in Cable